The driver uses Pascal calling conventions. The interface for some functions in this driver have changed from similar functions with a different prefix in older versions of the driver. The best thing to do here is to make sure that the VIs and the dll are in the same folder. Message 4 of By placing these files in that folder in the instr.
|Date Added:||3 March 2016|
|File Size:||33.21 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
But I dont’ know how to start with. Most Active Hardware Boards: Another thing I need to do is to control the current and temperature of the Newport Modular Controller Model laser diodes. I would be grateful if someone can knock up a VI or point me in the correct direction. The driver has an extensive help index which becomes a lot more clear once you read the instrument manual and figure out what the functions correspond to. lsbview
There are several advantages to using the dll form of the driver. Here is the attachment. Message 1 of No all statistics defined in the help file were returned. Does anyone know what happens?
This aglient will be fixed in a future release of the firmware. It includes a Visual Basic include file. Agileent 7 of The message shows “An exception occurred within the external code called by a Call Library Node.
Agilent Technologies / Keysight Technologies 86142B
This readme file contains additional information concerning the driver. Agulent 6 of A list of contact information for other countries is available on the Agilent Technologies website at http: All of directory paths which follow assume Windows Message 10 of The only exceptions are that it does not have a soft front panel or a knowledge base file. When sending commands directly to the instrument be careful to set the format agiilent time as required.
If you could adjust the error handling wires in that way to where it passes through the call library function node and take a screen shot of the agillent that you are getting. Do not build the driver in the directory in which it was installed. I am using Labview 7. This driver does no contain examples as the other driver, but there are three steps in creating the application which are:. Vierror occurrs. It is built on top of, and uses the services provided by VISA.
Agilent | Agilent x VXIplug&play Instrument Driver Read Me
This might have corrupted Labview’s memory. Do I need to initialize, configure and close the instrment driver? Please consult the documentation before using functions in the new driver.
A higher level of support from Agilent Technologies for the compiled driver 3. The help file is in error.
PRES” caused function to not work properly. Added 2 new applications, Amplifier Test and Source Tests.
The driver uses Pascal calling conventions.